High and low temperature test chamber
in the utilization rate of market is very high, especially it has important effect on industrial production.
The stability of the high and low temperature test chamber and hot and humid alternations are all very good, so in the physics environment simulation testing when it can have very good actual effect.
Yet high and low temperature test chamber in peacetime applications also must pay attention to the maintenance.
In the application process needn't often starting refrigeration compressor, refrigeration compressor starting time interval is over 15 minutes;
Ensure excellent office environment natural ventilation and no dust, to maintain the refrigeration compressor surface clean, if the long-term application of no environmental test chamber, should be placed in the indoor space as clean as possible.
High and low temperature test chamber working voltage must be stable in the workplace, if a short circuit, will cause the power flow is too big or too small, long-term operation in this kind of natural environment can lead to tremendous devastation of refrigeration compressor, proposed application of high and low temperature test chamber regulated power supply connection.
Application of the high and low temperature test chamber of the natural environment regulation: no strong electromagnetic radiation source;
Pavement leveling, good natural ventilation, no flammability, flammable and explosive, corrosive gas and dust;
Humidity of the air: & le;
Summer application to open the central air conditioning)
Temperature: 5 ℃ ~ 30 ℃ (
What's the average temperature in 24 hours & le;
Machinery and equipment surrounding set aside moderate the maintenance of the interior space.
Besides application machine equipment, please read the article by heart before this & quot;
Commodity use manual & quot;
, then carry out all kinds of actual operation, the operation to prevent don't mistake.
High and low temperature test chamber in high temperature experiment box when the temperature is very high.
Just finished experiment or during the whole process, if you want to open the tail to be very careful, in case of burns.
Refrigeration unit work, please do not touch the exhaust air conditioning copper tube temperature is very high during the whole process running, to prevent burns.
High and low temperature test chamber in ultra-low temperature experiment box when the temperature is very low.
Just finished experiment or during the whole process, if you want to open the tail to be very careful, in case of cold.
Although machines has perfect countermeasures to prevent to get an electric shock, but still have to pay attention to, especially in automatic control system for home appliances, work situation, avoid by all means touching part of household appliances.
Install security information content.
High and low temperature test chamber switch power supply used three-phase five wire, machinery and equipment four copper core cable, cable for three phase line, a line of zero.
Please idea zero difference.
Part of machinery equipment load and automatic control system for single phase electric, thrust fault zero may cause machine doesn't work and destroyed.
This machine wire connection regular column wire connection.
Ensure good machinery and equipment grounding device can make the machine more safety and.
Zero line and grounding line thrust fault may cause abnormal work machine equipment.
For customer site elements such as need to increase the cable, please make sure that commonly used line extension and the rating of the main parameters of switching power supply can consider goods of switch power supply requirements.
High and low temperature test chamber used in the whole process will cause a lot of heat.
Please install equipment in the district of natural ventilation is good.
Do not contact or near flammable materials or flammable natural environment application this machine equipment.
Don't be applied in magnetic field and apparent vibrations in the natural environment.
Application security information content.
Do not connect the sample switch power supply equipment of switch power supply, or it will improve machine equipment load switch power supply.
Unless it is machinery to set aside the sample power supply.
Ban the flammability, flammable and explosive, high erosion and strong radiation source objects.
Machinery and equipment to connect power circumstances or do not deliver, maintenance during the operation process.
High and low temperature test chamber, way and measure the metrological verification of the new project.
High and low temperature test chamber is the natural environment of each manufacturing industry must experiment equipment, the key for detection and clear electronic electrician, raw materials and goods in high temperature and cryogenic or stability experiment after the temperature of the environmental damage of main parameters and characteristics.
The interpretation of the above is to high and low temperature test chamber and maintenance instructions, hope can help you, if you want to learn more about information of high and low temperature test chamber, welcome to the online consulting service or call the company service hotline (
Website top right corner)
For consultation, we will wholeheartedly provide high quality service for you!
Tags: & nbsp  high and low temperature test chamber
In the present growing world of emerging technology, the has demanding operation in various sectors like tensile tester manufacturers
, tensile tester manufacturers, tensile tester manufacturers and many other industries at tensile tester manufacturers levels of manufacturing and designing.
You can always ask the experts for the most appropriate tensile tester manufacturers textile testing equipment
for your specific needs. Find the best support, prices and other to textile testing equipment solutions at GESTER.
GESTER International Co.,Limited deems that we can drive consumer transactions using high-tech tools like artificial intelligence and cognitive data sets.