When everyone is some application of laboratory equipment for the test
, sometimes encounter some problems.
In the constant temperature and humidity test chamber
temperature instability is a common problem, met a people first to analyzing a what conditions will appear this kind of problem.
1, first check the refrigeration compressor is in operation in the whole process can be started, if the operation can be started, means that from the main switch power supply to the constant temperature and humidity test chamber compressor electric line are all normal, no problems at that time can remove electrical system.
It's no problem for me to check the electrical system, the refrigeration system to check again.
2, check the a2 group refrigeration compressor unit, if the ultra-low temperature (
Stage compressor exhaust pipe and the suction pressure of a standard is a bit low, and inspiratory pressure in time, this suggests that the amount of refrigerant refrigeration unit is not significantly.
Unit 3, use hand to touch the main R23 compressor exhaust and suction pipes, such as found that the temperature of the exhaust pipe is not high, the temperature of the suction pipe is not low,
, it suggests that the lack of the main unit R23 refrigerant.
4, a leading unit and another for auxiliary units, high cooling speed, unit 2 groups also work at the early stage of the temperature to maintain link, 2 groups still another work unit.
For basic stable temperature, stop work auxiliary units, by the main unit to keep the temperature cool and smooth.
5, if the main unit R23 leakage, can make the cooling effect of the main unit is not large, because the cooling process, the two units in other work, so without temperature stable condition, and reduce the cooling speed.
6, in the temperature maintaining link, once the auxiliary units to stop working, the main unit and no cooling effect, constant temperature and humidity test the gas temperature in the cabinet will be slow to rise, when the temperature rises to a certain level, the automatic control system will start again auxiliary unit to cool, then auxiliary units and stop working, and so constantly, then there will be a low temperature to maintain the common fault condition.
The interpretation of the above is the cause of the constant temperature and humidity test chamber temperature instability, hope can help you, if you want to learn more about information of constant temperature and humidity test chamber, welcome to the online consulting service or call the company service hotline (
Website top right corner)
For consultation, we will wholeheartedly provide high quality service for you!
Tags: & nbsp  constant temperature and humidity test chamber
GESTER allocates customer service resources to the platform where their customers are most vocal.
We are proud to be one of the largest suppliers out there. You will love what we offer for your textile testing equipment
solution. Check our website at GESTER or call to talk to our customer service department with any questions you may have.
textile testing equipment are raising the stakes of social marketing, but they also ease the sales process by providing ways for tensile tester manufacturers to effectively interact with customers.