As a type of instrument often tried by textile enterprises, textile instrument
s are frequently tried in daily production. The textile instruments will inevitably lead to wear and failure during the trial process. How should people face and maintain some common faults? ? Returning to the factory for maintenance will waste a lot of time for the enterprise. The engineer will briefly introduce some methods. 1. Observation method Use vision, smell and touch. Sometimes, damaged components will change color, blistering or burnt spots; burned components will produce some special smell; short-circuited chips will be hot; virtual soldering or desoldering can also be observed with the naked eye . 2. Percussion and hand pressure method When we use the instrument, we often encounter the phenomenon of good and bad when the instrument is running. Most of this phenomenon is caused by poor contact or weak welding. For this situation, percussion and hand pressure can be used. The so-called 'knock' is to tap the plug-in board or component lightly with a small rubber squeeze head or other knocking objects to see if it will cause an error or shutdown failure. The so-called 'hand pressing' means that when a fault occurs, the components, plugs and sockets that are inserted after the power is turned off are pressed firmly by hand, and then the power is turned on to try whether the fault will be eliminated. If you find that it is normal to tap the case, and then it is not normal, it is best to re-plug all the connectors and try again. If it is not successful, you have to find another way. Third, the comparison method requires two instruments of the same model, and one of them is operating normally. To use this method, you must have the necessary equipment, such as a multimeter, an oscilloscope, and so on. According to the nature of comparison, there are voltage comparison, waveform comparison, static impedance comparison, output result comparison, current comparison, etc. The specific method is: let the faulty instrument and the normal instrument run under the same conditions, and then detect the signals at some points and then compare the two sets of measured signals. If there are differences, you can conclude that the fault is here. This method requires maintenance personnel to have considerable knowledge and skills. Fourth, the elimination method The so-called elimination method is to determine the cause of the fault by plugging in some plug-in boards and components in the machine. When a plug-in board or device is removed and the instrument returns to normal, it means that the fault has occurred there. 5. Capacitor bypass method When a certain circuit produces a strange phenomenon, such as a display disorder, the capacitor bypass method can be used to determine the part of the circuit that is probably malfunctioning. Connect the capacitor across the power and ground terminals of the IC; connect the transistor circuit across the base input or collector output, and observe the effect on the failure phenomenon. If the input terminal of the capacitor bypass is invalid and the fault disappears when the output terminal is bypassed, it is determined that the fault has occurred in this stage of the circuit. 6. Replacement method Two instruments of the same model or sufficient spare parts are required. Replace a good spare part with the same component on the faulty machine to see if the fault is eliminated. Seven, temperature rise and fall method Sometimes, the instrument will work for a long time or when the working environment temperature is high in summer, it will malfunction. Turn it off and check it is normal, stop for a while and turn it on again. This phenomenon is due to the poor performance of individual ICs or components, and the high temperature characteristic parameters fail to meet the index requirements. In order to find out the cause of the failure, the heating and cooling method can be used. The so-called cooling is to wipe anhydrous alcohol with cotton fiber on the part that may be malfunctioning when a malfunction occurs to cool it down, and observe whether the malfunction is eliminated. The so-called warming is to artificially raise the ambient temperature, for example, use an electric soldering iron to put it close to the suspicious part (note that the temperature must not be raised too high to damage the normal components) and try to see if the fault occurs. Eight, shoulder-riding method The shoulder-riding method is also called the parallel method. Put a good IC chip on the chip to be inspected, or connect a good component (resistance capacitor, diode, triode, etc.) in parallel with the component to be inspected, and keep good contact. If the fault is caused by an open circuit or Reasons such as poor contact can be ruled out by this method. Nine, state adjustment method Generally speaking, before the fault is determined, do not casually touch the components in the circuit, especially the adjustable devices, such as potentiometers. However, if you take pre-reference measures (for example, make a position mark or measure the voltage or resistance value before touching it), you can still allow touching if necessary. Maybe the fault will be eliminated sometimes after the change. X. Isolation method The fault isolation method does not require the same type of equipment or spare parts for comparison, and it is safe and reliable. According to the fault detection flow chart, segmentation and encirclement gradually narrow the fault search range, and then with signal comparison, component exchange and other methods, the fault will generally be found quickly. More about Textile Instruments: http://www.standard-groups.com/TextileGarment/
GESTER can also foster research that is more useful and influential in society at large.
If you are ready to stop the problem of tensile tester manufacturers and go back to normal, contact us at GESTER Instruments. GESTER International Co.,Limited is ready to help you out.
When it comes to textile testing equipment tensile tester manufacturers, GESTER International Co.,Limited is the name to reckon with. Not only are they best, they are the most experienced as well and provide wide range of services as well as products at affordable prices. Find out more information on GESTER Instruments.
When you choose to buy instead of tensile tester manufacturers, the money you save may allow you to buy multiple other necessities, more than you had initially planned on buying.