Constant temperature and humidity test
chamber can be high and low temperature environment simulation on commodity inspection carried out the experiment, but if the temperature of the simulation of natural environment is not set up, on the inspection of the actual effect and accuracy has great harm, the high and low temperature of constant temperature and humidity test chamber is not set value on how to solve?
A high temperature, constant temperature and humidity test chamber for is not on experimental indexes, such as temperature of high temperature on the experimental temperature values, must check the household appliances in fault detection system software.
If in the condition of the temperature is slow, so you must query the wind the wind circulation system adjustment partition of the respiratory system open if everything is normal, no wind circulation system of motor running if everything is ok.
If it is a high temperature after arriving temperature overshoot is large, so it is necessary to set set the main parameters of the PID, if environmental test chamber after arriving at the set temperature is still growing, already high overheating maintenance value, then the temperature and humidity controller is out of the common faults, it must be moved around the control instrument.
Not two, constant temperature and humidity test chamber temperature of experimental index, appear this kind of situation should observe the temperature change of situation, look at ultra-low temperature drop slowly or temperature reaches a certain value after the warming trend.
If temperature drop slower, need perseverance cryogenic personal studio is dry before experiment, the sample (s) work in the room is too much, the total product can lead to work in the room of the wind may not be adequate circulation system, commodities have calorific value, above these two conditions will cause temperature decrease slowly in the experiment box.
Otherwise you need to check equipment refrigeration unit is out of the common faults and professional maintenance staff to must technology.
The above explanation is the high and low temperature reach the set value of constant temperature and humidity test chamber solution, hope can help you, if you want to learn more about information of constant temperature and humidity test chamber, welcome to the online consulting service or call the company service hotline (
Website top right corner)
For consultation, we will wholeheartedly provide high quality service for you!
Tags: & nbsp  constant temperature and humidity test chamber
GESTER International Co.,Limited is different from other companies as we provide timely and unique services to our respected clients.
GESTER International Co.,Limited is the major textile testing equipment provider. tensile tester manufacturers businesses need the right tools at their disposal in order to handle tensile tester manufacturers. GESTER is your best choice.
The first machine to produce tensile tester manufacturers, the tensile tester manufacturers textile testing equipment was invented in tensile tester manufacturers in tensile tester manufacturers by tensile tester manufacturers and was subsequently improved.
We believe in keeping the customers happy and providing them with textile testing equipment at a very competent price.
textile testing equipment continued to evolve to having strong manufacturers develop huge marketers and people came to value their opinions about what to buy.