Constant temperature and humidity test
chamber is an important equipment of enterprise quality inspection product temperature and humidity, but in the long run the whole process, some customers find that machinery and equipment in the short term of setting temperature, environment humidity value, the constant temperature and humidity test chamber
why a cooling humidifying difficult?
Constant temperature and humidity test chamber in slow cooling humidification conditions, can endanger the actual effect and precision of test, be sure to carry out the solution of the corresponding treatment: 1, the temperature drop down: first to analyze this kind of situation is constant temperature and humidity test chamber is temperature drop slowly, or the temperature dropped to a certain value the development trend of temperature has warmed.
For temperature drop slowly, must check before the trial is dry personal studio, work in the sample (s) experiment is placed inside the room too much, cause the wind inside the room may not be sufficient in the circulatory system.
Above if no problem, want to consider is the main problem of refrigeration units, this kind of situation proposed contact manufacturers produce technology professional service personnel to carry out inspection maintenance.
2, environment humidity drop down: as the humidifying experiment, under the condition of constant temperature and humidity test in the environment humidity until not everyone set of values, or experimental box environment humidity value than the set value is much lower, the first to examine the dry bulb temperature sensor is less water in the stainless steel sink, stainless steel sink in the water level automatic control system is composed of a liquid level controller, liquid level controller is water supply equipment power supply all normal, liquid level controller is normal work.
If it weren't for above problems caused, should check the household appliances in automatic control system are all can't work normally cause can't humidifying experiment box.
The interpretation of the above is the constant temperature and humidity test chamber a cooling humidifying difficult reason, hope can help you, if you want to learn more about information of constant temperature and humidity test chamber, welcome to the online consulting service or call the company service hotline (
Website top right corner)
For consultation, we will wholeheartedly provide high quality service for you!
Tags: & nbsp  constant temperature and humidity test chamber
For the study, researchers defined GESTER as strategies to foster some social good, including programs that benefit community engagement, diversity, the environment, human rights and employee relations.
Above all, we expect to be a credit to the communities we serve, a valuable resource to our customers, and a place where our dedicated textile testing equipment can grow and prosper.
GESTER International Co.,Limited have found that nurturing relationships with clients by welcoming them to our factory can be valuable to all parties.
GESTER International Co.,Limited offers the best for indoor as well as outdoor use. To find your ideal at attractive offers, visit us at GESTER.
The proprietor has many years experience in providing promotion services and is a sought after expert in textile testing equipment.