ing machine is used to do mechanical properties such as tensile, compression of the sample test instrumentation.
Tensile testing machine is hard to avoid can appear in the application of some common faults, in the whole process of maintenance, we not only to master the proper maintenance way, aimed at the problem of maintenance process should pay attention to when also nots allow to ignore.
You know what are the decision factors influencing the tensile testing machine precision?
Simple introduce for everybody today.
1, screw: tensile testing machine screw gap is too large, or if RunHua insufficient, make to test machine data information is not accurate;
2, tensile testing machine transmission device: the sale on the market at present stage is generally more or synchronous belt transmission gear reducer.
If less than RunHua reducer, it may also affect the precision of the test;
3, sensors: tensile testing machine sensor, feature is stable is to determine the fundamental factors of tensile testing machine precision;
4, power plant, the speed of the motor rotation speed impact tensile testing machine, the rate of course affect the results.
Tensile testing machine multi-use of flexible packaging regulation, namely in the allocation of different jig, basically can do tensile, compression, bending, tear, cut, 180 & deg;
From, 90 & deg;
From the experiment.
At present there are some high-end tensile testing machine in addition to the above new project, because of its high precision sensor (
Some do one over two hundred and fifty thousand)
Can also detect frictional resistance.
Tensile testing machine for accuracy of ball screw is one of the biggest, but must give full play to the characteristic of the servo control system by computer practical operation can give full play to, and a full set of price is more expensive.
Selects the general screw and trapezoidal screw can the accuracy of the soft packaging regulations is 0.
In front of the transmission system is divided into gear and chain, is also a kind of relatively expensive, but used to high precision;
Behind a cost-effective, can be used to lower precision.
Sensor is a critical cost depends on its service life, in which optical magnetic induction is more excellent technical and general use above one hundred thousand times.
The interpretation of the above is the cause of the tensile testing machine precision, hope can help you, if you want to know more information about rolling tension tester, welcome to the online consulting service or call the company service hotline (
Website top right corner)
For consultation, we will wholeheartedly provide high quality service for you!
Tags: & nbsp  tensile testing machine
GESTER International Co.,Limited is willing to push up our sleeves and wade into unknown territory with equal parts fearlessness and optimism.
GESTER International Co.,Limited is a reliable company that offers wonderful . In addition, the company also provides related components to make it more efficient. To know more, go to GESTER.
GESTER International Co.,Limited always focus on the situation of global market and understands the importance factors of manufacturing textile testing equipment.