Judgment method There are two methods for judging the leather wear resistance index, namely the end point method and the appearance change method. 1. End point method: The number of rotations required to grind off l mm leather is the thickness index of wear resistance, and the number of friction rotations is recorded as the index of wear resistance. 2. Appearance change method: After the test sample has been rubbed for a certain number of times, observe its appearance change, such as color change, degree of wear and other appearance changes as an indicator of wear resistance. New conventional test methods for leather abrasion resistance: 1. Flat grinding method (1) Reference standard: 'GB/T 21196-2007 Martindale method of fabric abrasion resistance determination'. (2) Test equipment: There are many types of fabric flat grinding machines. At present, the Martindan tester
is the most used in the testing industry. Therefore, the flat grinding method is often referred to as the MARTINDALE friction testing machine method. (3) Test principle: The circular sample installed in the sample holder of the Martindale Abrasion Tester is carried out with the plane motion of the Lissajous figure and the abrasive (ie standard fabric) under a specified load. Friction, the sample holder can rotate freely around its axis perpendicular to the horizontal plane. Determine the wear resistance of the leather according to the total friction times of the sample damage. (4) Test method: During the test, the test sample of a certain size is in contact with the standard abrasive under the specified pressure, and the sample is moved relative to the abrasive in the Li Sharu track, so that the test sample is uniformly worn in multiple directions. 2. Taber Abrasion Method (1) Reference Standard: 'GB/T 2726-2005 Determination of Abrasion Resistance in Physical and Mechanical Tests of Leather'. (2) Test instrument: TABER wear resistance tester (3) Test principle: pre-grind 25 revolutions, put the tested sample on the level platform of TABER wear tester and rotate, and the two grinding wheels are given specific pressure Pressing on the test piece is optional, the axis of the grinding wheel is parallel to the horizontal plane, one grinding wheel is facing outwards, and the other is facing inward. In a certain period of time, all changes of the test piece are recorded. (4) Test method: Use a cutter to cut a sample with an outer diameter of 108mm and an inner diameter of 8mm. Then stick a hard sticker of the same size on the back of the sample. Place the center hole of the sample on the rubber pad of the screw, cover the pad, and fix the sample with a compression nut. Then put on the fixing ring, and tighten the fixing ring screw with a wrench. Install two H18 grinding wheels on the support arm. Each grinding wheel applies a force of 250±10g to the sample to be tested. Make sure that the side with the label is facing outwards so that the weight of the grinding wheel is 500g. Connect the vacuum cleaner to the vacuum cleaner. Connect the tube to the connector on the left side of the machine and start the vacuum cleaner. Set the number of tests according to the standard, and set the rotation speed to 72 revolutions. Then start the start button, turn on the instrument with a rotation speed of 60±5 times/min, rotate it horizontally, and perform the test according to the required number of times. In the test, use a brush to remove the attachments on the sample surface, observe the wear of the test, grind the leather to the specified end point, and record the final number of revolutions. 3. STROLL abrasion resistance method: According to ASTM D 3886-1999 'Standard testmethod for abrasion resistance of textile fabrics (inflated diaphragm apparatus)Above, rub the sample with sandpaper with specified surface characteristics. After the specified number of rubbings, the degree of wear of the sample is evaluated by appearance. Operation steps: Place the sample on the rubber film in a flat state, then place the sandpaper on the abrasive plate, and make the contact head connected with the sandpaper flush with the surface of the sandpaper. Then apply 28 kPa of air pressure under the diaphragm and 454 g of pressure above the abrasive plate, and ensure that the air pressure control and the contact between the inflated sample and the loaded sandpaper are in a stable and balanced state. Start the instrument and test according to the number of revolutions set by the counter. After the test, remove the sample, check and record the wear condition of the sample, and use the gray sample card to evaluate the color change in the test area according to GB/T 250-2008 'Textile color fastness test-gray sample card for evaluating discoloration'. It is not meaningful to compare the test results obtained by testing different materials and using the same experimental method, but the conclusions drawn from these results are consistent. Therefore, any method can independently quantify the wear resistance of the material. In general, the material of natural leather is relatively soft, so its wear resistance is poor. Coated leather has a harder texture and is not easy to wear, so it has better wear resistance.
GESTER International Co.,Limited have now make a decision to enlarge our company in other countries.
GESTER International Co.,Limited seeks to lead the industry by instilling pride in our customers, creating value for the market and sharing responsibility around the world.
The more people who do a certain thing, the more likely others are to do it as well. When GESTER can demonstrate their popularity or satisfaction across a wide customer base, other consumers are more likely to buy in as well.
GESTER International Co.,Limited's textile testing equipment are sturdy, easy to operate, friendly work machines that deliver high-quality tensile tester manufacturers for tensile tester manufacturers purposes.